Calibration is the process that scales undergo to ensure that they perform efficiently by giving precise and accurate results at all times. It needs to be done by skilled personnel.
Floor scales should be calibrated regularly depending on the workplace, how often it is used, and other factors like tear and wear.
Using a floor scale that is not calibrated can cause losses of money and time which tends to destroy the business’s image. Calibrating floor scales ensures that it is well maintained and you can identify signs of deterioration, and the proper preventive measures are taken.
It saves time and money, boosts customers’ confidence, and you can avoid legal implications. When the weighing instrument is not loaded, it must display a zero reading.
Floor scales need to be maintained and maintained regularly to keep their giving accuracy in weighing.
How Often Should You Calibrate Your Floor Scale?
Calibrating your floor scales will depend on the workplace surroundings, bulky offloading and loading, the recommendation of the manufacturers, and how often you use them.
When the floor scale generates output data with minor errors and within set tolerance limits, it is considered fit for usage.
Regularly used floor scales are quicker to undergo wear and tear as the accuracy diminishes over time, and it would be best to calibrate monthly.
It would be best if you seek advice from professionals like the floor scale manufacturers since they are well-versed and know when floor scales should be calibrated.
The workplace surroundings need to be favorable enough for the floor scales to test accurate weights.
Experts always recommend using well-calibrated weighing scales to eliminate risks associated with faulty readings.
To avoid the risk of inaccurate readings, experts usually recommend using properly calibrated weighing scales.
How to calibrate Floor scales
For safe operation, floor scales should be appropriately placed on a flat surface with enough surrounding space, free from any air current.
Turn on digital weighing scales a few minutes before calibrating them.
Ensure that the top of the platform is clear. It needs to be free from any objects.
Connect the floor scale indicator to the clear platform using a connection cable.
Place turn on the indicator that has been connected to the platform and give it time to configure.
Begin the calibration mode sequence by following the scale’s display instruction.
Once the calibration process has been completed, remove the calibration weight and return to normal weighing modes. The indicator will show the desired weight currently on the platform.
|Issue: Unstable weight display
Possible cause: load cells malfunctioning
Quick solutions: service call
Issue: Weight display not returning to zero
Possible cause: rough surfaces
Quick solutions: floor scales should be placed on an even ground.
Issue: inaccurate weights
Possible cause: obstacles
Quick solutions: regularly clean the floor scales and ensure it is free from contaminants.
Issue: unstable weight display
Possible cause: excessive vibrations
Quick solutions: use shock-absorbing materials.
It is possible to measure accurately, maintain regulatory compliance, and save money by performing routine scale calibration and adjustment.
The performance of floor scales is determined by calibration. Preventive maintenance visits should be scheduled based on how often the equipment is used and how much risk there is in the process.