300kg Load Cell | Beam Load Cells
- Accuracy: up to C6 load cells
- Sturdy aluminum alloy; stainless steel
- Different measure capacities are available for up to 300kg
- Dimensions and structures to your orders
- Simple and convenient installation
- Maximum platforms: 600mm x 600mm
- Protecting rate: IP 65, IP67, IP68
- Standard: ISO9001 CE&RoHs
CUBTE 200kg load cells are available in different dimensions and structures. As per your specific applications, you can choose 300kg single point load cells, 300kg s-type load cells, and 300kg bending beam load cells. etc for Bench scales, packaging and grading machines, bag filling equipment, and bottle filling machinery where accurate measurements are required.
◆Measure range: 300kg
◆Strain Gauge Sensor Type
◆IP65 Protection Rating
◆Accuracy class: C3 Multi-Range
◆Platform size: 800x800mm
◆Off-center load compensation up to 0.0233 %
◆Wire Size: 1.5 m ~ 12 m
◆Wiring Way:red = Exc+, green = Sig+, black = Exc-, and white = Sig-
◆Range: 50, 100, 150,200, 250kg, 300kg as per your needs
◆Type: OIML approved load cells
◆Protection Rate: IP68
◆Accuracy: C3 multi-ranage
◆Up to 200% Safe Overload, Linearity 0.05%
◆Stainless Steel structure
◆Low Power Consumption, Low Hysteresis
◆Range: 50, 80,100, 150, 200, 300kg varying capacities
◆Type: tension and compression load cell
◆ S-beam load cell for platform scales
◆Made of aluminum alloy
◆Accuracy: C3 ( +-0.03% class)
◆Sealed to IP67
◆comes with a full range of mounting accessories
◆Measuring range(Optional): 50kg, 100kg, 150kg, 300kg
◆Hermetically welded seal to IP68
◆constructed of high-quality aluminum
◆platform sizes up to 600 x 600mm
◆Cable Length: 6mtrs
◆Safe Overload:150% F.S
350 ± 50Ω
Compensated Temperate Range
-30 ~ +70 ℃
|Temperature Effect on Span||
± 0.014 %FS/10℃
Electrical Wire Connection
|4 color wire, Ø 5 × 110~500mm|
In addition to (300kg) load cells, we also manufacture and supply floor scales, axle scales, bench scales, crane scales, truck scales, pallet jack with scales, weighing indicator displays, and other weighing equipment hardware
How to simply tell if the load cell is damaged on spot?
The deformation of the strain gauge is very small/tiny. During installation and use, if the external force is removed, the original shape of the strain gauge cannot be restored and plastic deformation occurs. It indicates that the load cell is damaged.
What is the correct installation method of the single point load cell?
One end of the sensor is to be fixed through a screw. The other end remains suspended. Apply gravity in the direction indicated on the arrow. One important thing is that you must not directly press the white-coated part. So as not to damage the internal strain gauge of the sensor.
What size screws fit threaded holes on this type of 500g load cell?
M3 is ok.
What is the mass of the 500g load cell?
Single gross weight: 0.300 kg